Description: The Relativistic Heavy Ion Collider (RHIC) consists of two rings with cryogenic magnets at a 4.5K operating temperature. Control of positions of the dipole and quadrupole cold masses (iron laminations) and the beam position monitors (BPM`s) during production and installation is presented. The roll of the dipoles is controlled by a combination of rotating coil measurements with the surveying measurements. The center of the quadrupole magnetic field is obtained by direct measurement of the field shape within a colloidal cell placed inside the quadrupoles. Special attention is given to the triplet quadrupole alignment and determination of the field center position.
Date: May 1, 1995
Creator: Trbojevic, D.; Cameron, P. & Ganetis, G.L.
Item Type: Refine your search to only Article
Partner: UNT Libraries Government Documents Department