Description: This paper discusses the criteria that have been adopted tooptimize the signal processing in a shower detector to be employed as LHCbeam luminosity monitor. The original aspect ofthis instrument is itsablility to operate on a bunch-by-bunch basis. This means that it mustperform accurate charge measurements at a repetition rate of 40 MHz. Thedetector must withstand an integrated dose of 100 Grad, that is, two tothree orders of magnitude beyond those expected in the experiments. Tomeet the above requirements, an ionization chamber consisting of severalgaps of thickness 0.5 mm, filled with a gas that is expected to beradiation resistant, has been designed. Crucial in the development of thesystem is the signal processing, as the electronic noise may set thedominant limitation to the accuracy of the measurement. This is relatedto two aspects. One is the short time available for the chargemeasurement. The second one is the presence of a few meter cable betweenthe detector and the preamplifier, as this must be located out of theregion of highest radiation field. Therefore the optimization of thesignal-to-noise ratio requires that the best configuration of the chambergaps be determined under the constraint of the presence of a cable ofnon-negligible length between detector and preamplifier. The remoteplacement of the amplifying electronics will require that the front-endelectronics be radiation hard although to a lesser extent than thedetector.
Date: November 5, 2001
Creator: Datte, P.S.; Manfredi, P.F.; Millaud, J.E.; Placidi, M.; Ratti,L.; Speziali, V. et al.
Item Type: Refine your search to only Article
Partner: UNT Libraries Government Documents Department