Description: Previous work on the detection of mercury using the cavity ring-down (CRD) technique has concentrated on the detection and characterization of the desired mercury transition. Interferent species present in flue gas emissions have been tested as well as a simulated flue gas stream. Additionally, work has been done on different mercury species such as the elemental and oxidized forms. The next phase of the effort deals with the actual sampling system. This sampling system will be responsible for acquiring a sample stream from the flue gas stack, taking it to the CRD cavity where it will be analyzed and returning the gas stream to the stack. In the process of transporting the sample gas stream every effort must be taken to minimize any losses of mercury to the walls of the sampling system as well as maintaining the mercury in its specific state (i.e. elemental, oxidized, or other mercury compounds). SRD first evaluated a number of commercially available sampling systems. These systems ranged from a complete sampling system to a number of individual components for specific tasks. SRD engineers used some commercially available components and designed a sampling system suited to the needs of the CRD instrument. This included components such as a pyrolysis oven to convert all forms of mercury to elemental mercury, a calibration air source to ensure mirror alignment and quality of the mirror surfaces, and a pumping system to maintain the CRD cavity pressure from atmospheric pressure (760 torr) down to about 50 torr. SRD also began evaluating methods for the CRD instrument to automatically find the center of a mercury transition. This procedure is necessary as the instrument must periodically measure the baseline losses of the cavity off of the mercury resonance and then return to the center of the transition to accurately measure the …
Date: June 30, 2003
Creator: Carter, Christopher C.
Item Type: Refine your search to only Report