Description: This Engineering Note studies the design of the Dry Air Purge that is going to flow through the Warm Tube of the End Calorimeter of the D-O Calorimeter. The Tev tubes through the E.C. can be thought of as a cluster of concentric tubes: The Tev tube, the warm (vacuum vessel) tube, 15 layers of superinsulation, the cold (argon vessel) tube, and the Inner Hadronic center support tube. The Dry Air Purge will involve flowing Dry Air through the annular region between the Warm Tube and the Tev Beam Pipe. This air flow is intended to prevent condensation from forming in this region which could turn to ice under cryogenic temperatures. Any ice formed in this gap, could cause serious problems when these tubes are moved. The Air will flow through a Nylon Tube Fitting -1/4-inch I.D. to 1/8-inch male pipe thread (Cole Palmer YB-06465-15) see Drawing MC-295221 (Appendix A). This fitting will be attached to the Nylon 2-inch Tube-Wiper and Seal Assembly which is clamped to the ends of the Warm Tube (Appendix A). This note includes drawings and calculations that explain the setup of the Dry Air Purge and give the required information on the pressure drops through the setup. The Equations and properties used in the calculations were obtained from the Applied Fluid Dynamics Handbook by Robert D. Blevins and Fluid Dynamics Second Edition by Frank M. White.
Date: August 14, 1991
Creator: Leibfritz, J. R.