Description: The South endcap calorimeter vessel was moved into Lab A on Sept. 18, 1990. A cooldown of the pressure vessel with liquid nitrogen was performed on Sept. 26 to check the vessel's integrity. With the pressure vessel cold, the insulating vacuum was monitored for leaks. Through out the testing, the insulating vacuum remained good and the vessel passed the test. The cold test was carried out per the procedures of D-Zero engineering note 3740.220-EN-250. The test was very similar to the cold test performed on the Central Calorimeter in October of 1987. The test of the ECS was performed in the same manner using the same equipment as the ECN cold test. Reference D-Zero engineering notes 3740.210-EN-122, 3740.000-EN-I07, and 3740.210-EN-II0 for information about the CC cold test. Reference EN-260 for the results of the ECN cold test. The insulating vacuum space was pumped on while equipment was being connected to the pressure vessel. Two hours after starting to pump with the blower the vacuum space pressure was at about 40 microns. The pumping continued overnight (another 16 hours). In the morning the pressure was 11.5 microns. A rate of rise test was performed. With the pump valved off, the pressure rose to 14 microns within 5 minutes and then rose to 16 microns in 6 hours (0.33 microns/hour). After all connections were made to the pressure vessel, a vacuum pump with an estimated effective pumping speed of about 70 scfm was valved on. After 18 hours, the pressure vessel was down to 270 microns. An additional day of pumping took the pressure down to only 250 microns. A leak was then found and fixed around the seal of the rupture disc. The pump was put on line again. The pressure vessel with pump on line was 27 microns after 16.5 ...
Date: November 26, 1990
Creator: Rucinski, R.
Item Type: Refine your search to only Report