Heat treating and inspection of metals Page: 39
The following text was automatically extracted from the image on this page using optical character recognition software:
HEAT TREATING AND INSPECTION OF METALS
b. The machine (fig. 6) consists principally of a base, or standard,
elevating screw, anvil, penetrator, device for applying the load, and
3 7 1
Penetrator. 7. Gage sleeve.
Weight pan. 8. Indicating gage dial.
Anvil. 9. Crank handle.
Elevating screw. 10. Plunger rod.
Hand wheel. 11. Major load.
Indicating gage stem.
FIGURE 6.-Rockwell hardness tester.
(1) The dial indicator is provided with two scales. The black or
C scale and the red or B scale. The C scale is used when testing with
the diamond cone and 150 kilogram load while the B scale is used in
connection with the steel-ball penetrator and 100 kilogram load.
Readings, when recorded, must be prefixed by the letter B or C to show
which scale has been used and a column is provided for each scale in
table VII. When the readings fall below the table value, C-20 (B-98),
Here’s what’s next.
This book can be searched. Note: Results may vary based on the legibility of text within the document.
Tools / Downloads
Get a copy of this page or view the extracted text.
Citing and Sharing
Basic information for referencing this web page. We also provide extended guidance on usage rights, references, copying or embedding.
Reference the current page of this Book.
United States. War Department. Heat treating and inspection of metals, book, September 10, 1941; Washington D.C.. (digital.library.unt.edu/ark:/67531/metadc96657/m1/41/: accessed January 21, 2018), University of North Texas Libraries, Digital Library, digital.library.unt.edu; crediting UNT Libraries Government Documents Department.