METHOD FOR DETERMINATION OF LIQUID DENSITY AND VISCOSITY OF ORGANIC COOLANTS OVER THE TEMPERATURE RANGE (M.P. TO 850 F) Page: 4 of 15
This report is part of the collection entitled: Office of Scientific & Technical Information Technical Reports and was provided to Digital Library by the UNT Libraries Government Documents Department.
The following text was automatically extracted from the image on this page using optical character recognition software:
WRITTEN BY A.TDR N0.
ERIN G. Asanovich ATOMICS INTERNATIONAL 3863
Organic Reactors A Division of North American Aviation, Inc.
UNIT Physical and TECHNICAL DATA RECORD LEDGER ACCT.
APPROVED BYORA SUB-ACCT. 46
- ______ ________TWR ____
Physical Properties of Hydrocarbons PAGE
1 OF -12
T0: R.H.J. Gercke
COPIES TO: OMRE-Advanced OMR Distribution
Piqua OMR Distribution
G. Asanovich R.T. Keen F.C. Silvey
R. Balent W.R. Martini C.A. Trilling
K.H. Campbell W.E. Parkins Group 796-61 (15)
R.J. Edwards M. Silberberg
SUBJECT: Method for Determination of Liquid Density and Viscosity of Organic
Coolants over the Temperature Range (M.P. to 850*F)
1 STATEMENT OF PROBLEM . . . . . . . . . . . . . . . . . . . PAGE 1
II SUMMARY OF RESULTS.AND RECOMMENDATIONS ....... . . . . . . . PAGE 1
III METHOD USED, DESCRIPTION OF EQUIPMENT, SAMPLE CALCULATIONS * PAGE 2 - 9
IV REFERENCES AND APPENDICES . . . ... . . . . . . . . . . . PAGE 9 - 12
I STATEMENT OF PROBLEM
To measure the liquid density and viscosity of organic coolants over
the temperature range 300 to 850*F.
A knowledge of the density and viscosity of organic coolants as a
function of temperature and extent of decomposition is important to
reactor operation. Techniques were developed to measure these properties
over the temperature range 300 to 850*F.
Density is determined as a function of temperature bye measuring'
the volume occupied by a known weight of sample in a modified Lipkin
pycnometer. The pycnometer is pressurized with an'inert gas to prevent
boiling and maintained at thermal equilibrium in a hot air furnace.
Estimated accuracy of measurements is 0.3 per cent.
Viscosity is determined as a function of temperature Iby measuring
efflux times in a (Cannon-Manning) modification of the Ostwald semi-
micro viscosimeter. The Ostwald pipet is.pressuried with an inert gas
to prevent boiling. Estimated relative accuracy of'rreasurements is
3 per cent.
S C3 0
Here’s what’s next.
This report can be searched. Note: Results may vary based on the legibility of text within the document.
Tools / Downloads
Get a copy of this page or view the extracted text.
Citing and Sharing
Basic information for referencing this web page. We also provide extended guidance on usage rights, references, copying or embedding.
Reference the current page of this Report.
Asanovich, G. METHOD FOR DETERMINATION OF LIQUID DENSITY AND VISCOSITY OF ORGANIC COOLANTS OVER THE TEMPERATURE RANGE (M.P. TO 850 F), report, May 14, 1959; Canoga Park, California. (digital.library.unt.edu/ark:/67531/metadc1026366/m1/4/: accessed October 21, 2018), University of North Texas Libraries, Digital Library, digital.library.unt.edu; crediting UNT Libraries Government Documents Department.